|Zero temp. coefficient||-40～85°||±0.4||°/℃|
|EMC||According to EN61000 and GBT17626|
|Insulation resistance||≥100 megohm|
|Anti-impact||100g@11ms,three axes direction(half sinusoid)|
|Housing material||Sand-grinding oxidation of aluminum alloy|
|Screw||Three M4 screws|
|Cable||Standard 1-meter length, anti-ultraviolet, shielded cable 4*0.4mm2|
|Weight||60g (exclude cable)|
INSTRUCTIONS FOR USE
1. This tilt sensor measures the tilt angle of the object through the principle of sensing the gravity of the earth. When installing, try to ensure that the direction of the sensor axis of the sensor is parallel to the direction of the tilt axis of the measured object to achieve the best measurement accuracy. It must be tight, flat, and stable. If the mounting surface is uneven, it will easily cause errors in the measurement angle of the sensor.
2. The inclination sensor can be installed and measured arbitrarily on six sides. After the installation is complete, use the zero setting function of the sensor to set the current position to zero. In the internal memory of the product, after zeroing, the product will work with the current position at zero degrees.) The setting method is as follows: short-circuit the sensor setting wire (gray) and the ground wire (black) for more than 3 seconds, and the sensor power indicator will turn off until the indicator light flashes again, then release the setting line, the zero setting is completed, and the indicator light returns to the constant light working mode.
3. The protection level of this sensor is IP67. Rain or strong water spray will not affect the operation of internal devices. Please do not immerse it in water for a long time to avoid damage to the internal circuit of the product. The manufacturer will provide paid maintenance service.
4. After the product installation is completed, please pay attention not to short-circuit the signal line and the positive pole of the power line to avoid burning the output circuit. Since the negative signal of this product and the negative pole of the power supply are shared, please connect the negative signal of the collection end and the negative pole of power supply of the product.
|Contact Person :||sean|